D CV. Read this owner’s manual thoroughly before use Owner’s operation manual for series. DT Series. 3 1/2 Digital Multimeter. Read this owner’s manual thoroughly before use. DT series multimeter have been designed according to IEC concerning electronic measuring. View and Download Histest DT owner’s operation manual online. DT Multimeter pdf manual download.
|Published (Last):||17 December 2012|
|PDF File Size:||7.97 Mb|
|ePub File Size:||2.16 Mb|
|Price:||Free* [*Free Regsitration Required]|
Set the range switch to the desired range.
Front Panel Description This function allows the meter to output a signal as a signal generator. Before measuring current, check the meter’s fuses and turn power off to the circuit before connecting the meter nanuale the circuit. Enter text from picture: If the resistor to be measured is connected to a circuit, disconnect circuit power and discharge all capacitors before measuring resistance.
Before attempting to open the case, always be sure that test leads have manualw disconnected from measurement circuits. Table of Contents Add to my manuals Add.
In the latter case, the transient stresses are variable; for that reason, its necessary that the transient-withstand -capability of equipment is made known to the user. Set the range switch to ” ” or ” ” range. Page 6 To use safely, each measurement can not last for more than 10 seconds, and the interval between each measurement must be more than 15 minutes. To use safely, each measurement can not last for more than 10 seconds, and the interval between each measurement must be more than manhale minutes.
A test signal will appear between the and “COM” jacks.
Owner’s operation manual for DT Series 3 ½ Digital Multimeter
By pressing ‘print’ button you will print only current page. Safety Information If the defect has been caused by a misuse or abnormal operation conditions, the repair will be billed at a nominal cost.
Page of 13 Go. Use the proper terminals, function, and range for your measurements.
Histest DT830 Owner’s Operation Manual
If the defect has been caused by a misuse or abnormal operation conditions, the repair will be billed at a nominal cost. Comments maanuale this Manuals Your Name. Operating Instruction “COM” jack. Got it, continue to print.
Close case and tighten screws completely before using the meter to avoid electrical shock hazard. The voltage is approximate 5Vp-p. Related Manuals for histest DT No related manuals. Don’t have an account? A coupling capacitor should be used when connecting the meter to a circuit.
DT-830D Manual Ranging Digital Multimeter
Don’t show me this message again. This function allows the meter to output a signal as a signal generator. To print the manual completely, please, download it. Battery And Fuse Replacement Before attempting to open the case, always be sure that test leads have been disconnected from measurement circuits.
Disconnect circuit power and discharge all high-voltage capacitors before testing resistance, continuity, diodes, or capacitance.